Function Generator A / B Equipment Hewlett-Packard, HP; Palo Alto, CA, build , 8 pictures, 1 schematics, United States of America, tubes. I finally got around to purchasing my first function generator. It was won off Ebay and is in good shape. Everything appears to function except for. The HP A was introduced in the catalog as a compact voltage- controlled function generator with 10 decades of range. Available output waveforms are.
|Published (Last):||21 May 2015|
|PDF File Size:||20.14 Mb|
|ePub File Size:||4.25 Mb|
|Price:||Free* [*Free Regsitration Required]|
If any one power henerator goes bad everything gets screwed up. Learn more – opens in a new window or tab. Test equipment repair shops charge a arm and leg for any repair.
Evolution of Function Generators during the to Period. Each supply 10 volts to the generators circuit. In 3310, one output may be switched to supply an inverted version of the waveform appearing at the other output, enabling the instrument to supply push-pull signals balanced to ground on any of the waveforms at any frequency.
HP 3310A function generator repair??
Two of the power supplys show no voltage. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Geneator on Pinterest – opens in a new window or tab.
From bottom to top: HP A Function Generator powers on but indicator light does not work. Pulse bursts, useful for testing counting circuits, are obtained by using the multiple-cycle mode. Either output supplies any one of the three waveforms sine, square, or triangular independently of the other output and with amplitudes up to 35 volts peak-to-peak into an open circuit or 15 volts into Ohms.
What makes this restoration project particularly difficult is that the published schematics are incomplete. Being sold as is. The Model A Function Generator combines two signal sources in a single box to produce a wide variety of test signals from 0,01 Hz to 13 MHz. One way they should show open, the other way read somewhere between and ; that’s their Vf when a certain current is applied to them.
This item will be shipped through the Global Shipping Program and includes international tracking.
HP 3310A 5 MHz Function Generator Hewlett Packard
Funciton an output but seems to behave oddly. Let me know if you are interested and what you might charge. Controlling a Toaster Oven with a PICAXE – Part 2 Soldering surface mount components by hand can be done, but reflow soldering in an oven is much quicker and produces more consistent results.
Seller assumes all responsibility for this listing. Hooksett, New Hampshire, United States. I can see the caps from the top but will have to take the bottom cover off to really inspect them. Two of the caps are rated at 30 volts and the other two are rated at 50 volts. It has never been serviced because the unbroken manufacturers warrenty seal is still over a lower case screw head. Additional flexibility of operation is made possible by the provision for electronic frequency control.
This item will ship to United Statesbut the seller has not specified shipping options. Nov 9, Output amplitude over the full frequency range is 1 mV to 10 V p-p into 50 Ohms.
There is no “I” in Steve: Repairing HP A Function Generator
Please enter a number less than or equal to 1. I can’t tell from the manual whether the transformer is soldered to the board, or if it has a connector. Skip to main content. Is there a quick way to see if any part is good without unsoldering it from the circuit board? Each power supply has a large uf electrolytic capacitor in its circuit. An item that does not function as intended and is not fully operational. Nov 7, 9. Function Generator Gennerator by marims in forum: A “B” version of the was introduced in the catalog.
A FUNCTION GENERATOR [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
Practically all of them are missing 33310 least half of the information. No guts, no glory! Refer to eBay Return policy for more details. Function Generator Posted by Dr.